Find great deals for HP Hewlett Packard A LCZ Meter T Shop with confidence on eBay!. Agilent HP A LCZ Meter is a general purpose Impedance measuring instrument designed to measure circuit Components such as capacitors and inductors. The HP Yokogawa (Made in Japan) LCR meters are still very useful, these are % basic accuracy LCR meters – with various test frequencies.
|Published (Last):||3 December 2013|
|PDF File Size:||8.81 Mb|
|ePub File Size:||5.84 Mb|
|Price:||Free* [*Free Regsitration Required]|
My Background Visual Presentation. Leave a Reply Cancel reply Your email address will not 4276x published. They were generally very accurate and of high quality. PTS Frequency Generator.
Keysight Technologies (Agilent HP) A used or new for sale at used-line
This is a nice board, with several hybrids. Below is a shot of the meter measuring a little 2. This site uses Akismet to reduce spam.
Learn how your comment data is processed. Small Signal Averaged Results.
Goldstar FG Function Generator. In the corners and gaps, 4276s wasp, and a dried frog. Take care of the fan hybrid, which is a thin ceramic plate — overall, a rather delicate construction. A good phase — you clearly see the switching action of the transistor, grounding the phase wire for some time to move the fan forward.
HP 4276A LCR Meter: several issues, and several animals found
Clearly, something is wrong with the range switching. Thawing trays – some facts added to the internet BS. Save my name, email, and website in this browser for the next time I comment. Avg Inductor Current Ambiguity. And, you can apply DC bias as needed hhp test ceramic capacitors, diodes, etc. Tektronix D1 Curve Tracer.
HP A LCR Meter: several issues, and several animals found | SimonsDialogs
Seems the input amplifier is clipping the signal. Download an averaged model demo. HP A Pulse Generator. Some test — still not working — at least not in all ranges.
Fluke 45 Dual Display Multimeter.
Navigation Home Blog re This Website It arrived in good overall state and well packaged, but very dirty inside. The test setup — with an external 8 Volts supply. Incorporate saturating inductor effects realistically. Cleaned it thoroughly with an eraser, and alcohol. Notify me of follow-up comments by email. If You’re Unfamiliar With Averaging. It even had some traces of corrosion — maybe it was running hot or some other failure. Fluke A Differential Voltmeter. Add Avg-based Vout waveform.
General Radio K Decade Capacitor. I would this meter for a marginal price, in Japan. As a last measure, inspected all the X and Y rated capacitors — it good practice to do so with any old instrument. Back to Boatanchor Test Equipment.
Next — we have to fix the analog circuit. The main difference is that the A doesn’t measure DC resistance. Probably, this instrument has been stored in some shed? Large Signal Startup Results. I bought the 4 terminal Kelvin test leads separately, as the used eBay unit didn’t come with them. However, it can measure the capacitance and dissipation factor or ESR at spot frequencies from Hz to 20 kHz, and it does measure AC resistance.