The contents of this manual, including all graphics and photographs, are the property of Roche. Diagnostics. No part of this document may be reproduced or. Then, as necessary, refer to the User Manual: The CoaguChek XS System User Manual is a comprehensive guide to the meter and test strips. It is designed to. Please send questions or comments about this manual to your local Roche ACCU-CHEK, COAGUCHEK and SAFE-T-PRO are trademarks of Roche. All other.
|Published (Last):||5 May 2007|
|PDF File Size:||13.43 Mb|
|ePub File Size:||17.26 Mb|
|Price:||Free* [*Free Regsitration Required]|
Got it, continue to print. Get Ready Take a test strip out of the then press M container. Operating Conditions Operating Conditions Refer to the Getting Started guide to learn how to set up the meter and xx for and run your first blood-clotting-time test. The CoaguChek XS PT test strip may be used for patients under a combination therapy of oral anticoagulants plus heparin injections. For more information, see re-insert it.
ROCHE COAGUCHEK XS USER MANUAL Pdf Download.
Fill the Capillary Tube When the flashing test strip 4. To exit Erase mode without To erase all stored results: To print the manual completely, please, download coagychek.
Open Battery Compartment With the meter turned off, turn it over. The letters mem The CoaguChek XS Meter automatically stores up to test indicate that you are viewing a results and their dates and times in its memory.
This graph shows the mean bias for INR results; the overall mean bias was 0. Operating Conditions Refer to the Getting Started guide to learn how to set up the meter and prepare for and run your first blood-clotting-time test. The Coefficients of Variation presented for precision of whole blood duplicates indicate that the pre- cision of the CoaguChek XS monitor for both capillary and venous whole blood samples is very good.
If you see the another method. Enter text from picture: Don’t have an account? You may see the following error messages while using the The meter is too cold or too Turn the meter off and allow it CoaguChek XS Meter.
Test Strip Interference Solution: Repeat the test using a new test Error: The correlation was 0. View Most Recent Result The most recent test result appears. Keep the tube voaguchek, and allow it to fill by capillary action.
Page 24 For more information, see re-insert it.
Roche CoaguChek XS User Manual
Integrated Quality Controls Tips for a Good Fingerstick The CoaguChek XS System has quality control functions integrated For fingerstick blood testing, increasing the blood flow in the finger into the meter and test strips, so you never have to run quality will help you get a good drop of blood.
The correlation coefficient is 0. Code Chip Code Chip 2. Regularly check that all segments You can access these settings by going to Setup mode.
Preparing For A Test 1. Dose the Test Strip Discard the first 4 drops of blood.
Allow to Dry With the cover off, allow the The CoaguChek XS Meter includes advanced features that you can coaguchel strip guide to dry for set up as necessary. There is a potential risk of infection. Blood 1 hanging drop of blood may be collected directly using a butterfly or a needle and syringe. If you see an error message, first try to warm to measure correctly.
Page 92 Tables 1 and 2; a visual representation is provided in Graphs 1 and 2. Match Codes Make sure that the three- number code on the new test Each box of test strips comes with its own code chip.
Clean the meter whenever Cleaning manial Test Strip Guide it looks dirty or, if you prefer a regular schedule, clean the meter each time you open a new box of test strips. Table Of Contents User Manual Comments to this Manuals Your Name.
Table of Contents Add to my manuals Add. Integrated Quality Controls Integrated Quality Controls Tips mabual a Good Fingerstick The CoaguChek XS System has quality control functions integrated For fingerstick blood testing, increasing the blood flow in the finger into the meter and test strips, so you never have to run quality will help you get a good drop of blood. Tables 1 and 2; a visual representation is provided in Graphs 1 and 2.